4-1. Tools and Test Equipment Required
a. To insure that the test set is always ready for
Tools and test equipment required at organizational
operation, it must be inspected systematically so that
maintenance consists of:
defects may be discovered and corrected before they
a. T o o l K i t , E l e c t r o n i c E q u i p m e n t T K - 1 0 1 / G .
r e s u l t in serious damage or failure. The necessary pre-
b. T o o 1 K i t , E l e c t r o n i c E q u i p m e n t T K - 1 0 5 / G .
ventive maintenance checks and services to be per-
f o r m e d are listed and described in table 4-1. The item
d. T e s t S e t , E l e c t r o n T u b e T V - 7 / U .
numbers indicate the sequence of and minimum in-
spection required. Defects discovered during operation
o f the test set will be noted so that repairs maybe made
Remove dust and corrosion from metal surfaces by
w h e n test set operation has ceased. Stop operation im-
lightly sanding the surfaces with fine sandpaper.
mediately if a deficiency is noted during operation
B r u s h two thin coats of paint on the bare metal to pro-
which would damage the equipment. Record all defi-
tect it from further corrosion. Use the following
ciencies together with the corrective action taken on
references as a guide to repainting and refinishing
the appropriate maintenance form (TM 38-750).
a. S B 1 1 - 5 7 3 .
b. S o m e o f t h e i t e m s t o b e i n s p e c t e d a r e l o c a t e d i n
b. A R 7 4 6 - 5 .
the interior of the teat set and are only accessible by
C. T B 4 3 - 0 1 1 8 .
removing the panel assembly. Refer to paragraphs 4-7
ITEM TO BE INSPECTED
Secure the cover; make certain the latches and hinge work properly.
Inspect the exterior surfaces of the test set. They should be clean, free of dust, grease, and fungus.
pointers are neither broken nor bent and rotate freely and the POWER 16 lights when power is
blistered components; check for frayed, broken, or abraded wiring; check solder connections to in-
sure proper electrical conduction, and check for the accumulation of dust, dirt, or grime.
Test all tubes for proper emission and absence of internal shorts.
Check to see that the teat set is functioning properly by performing the procedure given in paragraph
Check DA Pam 310-4 to see that all publications are complete, serviceable, and current.
Check DA Pam 310-7 for applicable MWO's.
Check the test set to determine if MWO's have been performed. All URGENT MWO's must he ap-